British regulators issue a safety alert on a battery defect in Medtronic EnTrust defibrillators over concerns that a short circuit could leave patients without pacing.
British regulators issued a safety warning on Medtronic's (NYSE:MDT) EnTrust implantable cardioverter defibrillators over concerns that some of the devices may have a defect that could cause the batteries to deplete prematurely.
U.K. authorities recommended that patients with the affected devices check with their doctors and reprogram their devices to beep in case of low battery, Reuters reported.
The Minneapolis, Minn.-based med-tech giant issued an alert earlier this month on its EnTrust and and Escudo ICDs, warning that less than 1% of the global distribution of the devices may be susceptible to an internal short circuit that could drain the device battery without providing adequate warning.
"Typically, these heart devices have 3 months of normal operation after the device signals the need for replacement," Medtronic spokeswoman Kathleen Janasz told MassDevice in an email. "EnTrust devices may require replacement sooner than expected, and result in less than 3 months of normal operation once the device indicates the need for replacement."
About 69,000 patients have received the EnTrust device, according to the news service.
Edwards Lifesciences reports 3rd-quarter sales and earnings growth in excess of 20% and raises its...
A federal jury in Texas reportedly clears Johnson & Johnson's DePuy Pinnacle hip implant in the...
Federal auditors say the FDA needs to ramp up its cybersecurity efforts in order to deter hackers.
ReFlow Medical announced FDA clearance for its Wingman35 crossing catheter.
Sympara Medical quietly secures $5.5 million in funding from a handful of investors as it develops a...