British regulators issue a safety alert on a battery defect in Medtronic EnTrust defibrillators over concerns that a short circuit could leave patients without pacing.
British regulators issued a safety warning on Medtronic's (NYSE:MDT) EnTrust implantable cardioverter defibrillators over concerns that some of the devices may have a defect that could cause the batteries to deplete prematurely.
U.K. authorities recommended that patients with the affected devices check with their doctors and reprogram their devices to beep in case of low battery, Reuters reported.
The Minneapolis, Minn.-based med-tech giant issued an alert earlier this month on its EnTrust and and Escudo ICDs, warning that less than 1% of the global distribution of the devices may be susceptible to an internal short circuit that could drain the device battery without providing adequate warning.
"Typically, these heart devices have 3 months of normal operation after the device signals the need for replacement," Medtronic spokeswoman Kathleen Janasz told MassDevice in an email. "EnTrust devices may require replacement sooner than expected, and result in less than 3 months of normal operation once the device indicates the need for replacement."
About 69,000 patients have received the EnTrust device, according to the news service.
Covidien’s 3-year study results showing positive outcomes for its Urgent PC Neuromodulation system in...
NEC Display Solutions wins 510(k) clearance for its 21-inch medical monitor systems.
J&J's former head of global cardiology makes the jump to CEO at Itamar Medical
Sunshine Heart raises $1.5 million of a hoped-for $25.5 million equity funding round.
J&J subsidiary Lifescan asks the courts to ban Shasta from using its logo, saying problems with...